Get our latest news
The placement of the Triple beam balance should pay attention to environmental sanitation
The placement of the Triple beam balance
The Triple beam balance should be correctly placed in a safe weighing room or a stable workbench to avoid airflow fluctuations caused by environmental factors, large temperature changes, direct sunlight, vibration and static electricity. This is the most basic requirement for the placement conditions of a Triple beam balance. Usually in the inspection work, in some enterprise laboratories, it is often seen that the Triple beam balance is mixed with the drying box, the horse boiler, the vibration table, etc., which will cause the electronic three-beam balance to be unstable or unable to weigh in use.
Always maintain the sanitation of the interior of the Triple beam balance
At the time of the inspection of the enterprise, some laboratories are close to the workshop, and the coal powder or other dust generated in the production is scattered inside and outside the Triple beam balance. In addition, the operator accidentally spilled items in the three-beam balance room during the weighing process, resulting in weighing The result is inaccurate. Therefore, be sure to keep the weighing room of the triple beam balance clean. Once the items are scattered, they should be carefully cleaned with a soft cloth or brush in time. When weighing volatile and corrosive items, they should be placed in a closed container to avoid corrosion and damage to the electronic triple beam balance. It is strictly forbidden to impact and overload the weighing pan to avoid damage to the Triple beam balance. The temperature of the object placed in the triple beam balance should not be too high to avoid damage to the instrument. The general temperature is ≤70℃.